The CDB is a dual precision monostable multivibrator with independent trigger and reset controls The device is retriggerable and resettable and the control. B Monostable Multivibrator are available at Mouser Electronics. Mouser offers inventory, pricing, & datasheets for B Monostable Multivibrator. Data sheet acquired from Harris Semiconductor. SCHSE. Features. • Retriggerable/Resettable Capability. • Trigger and Reset Propagation Delays.
|Published (Last):||24 August 2011|
|PDF File Size:||1.91 Mb|
|ePub File Size:||18.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
Like I mentioned, I was having trouble with CD and seems the pinout in the multisim is quiet different with the datasheet for the CD and CD When only a single- or bi-axis measurement is needed at a datsheet, the small size and possibility of single axis powering supply makes Type B the right choice.
NTEB – IC-CMOS Dual Precision Monostable Multivibrator
If you found that they can be used as a sustitute I would think that there schematic dattasheet be very similar to get the response that you want, but the confirmation of the wiring is somehting that you can get from the devices datasheet. Would you be able to show me how to connect it?
This dwtasheet the diagram that I was working on. So were you able to find the workaround afterwards? Message 4 of I cannot see the attachment. Message 1 of So this is probably use less at this point, but in the most recent version of Multisim Are you trying to connect the CD using the diagram for the CD?
Which is the picture that you are using and what are the results that you are getting with it? Subscribe to Waves and get the stories from the world of sound and vibration. Message 3 of datsaheet Message 6 of Message 8 of Most Active Software Boards: I am glad to see that you found the documets helpful!
Designed to simplify your testing by covering most of the different needs of a modern test lab with one sensor. TEDS function and the possibility of mounting with an M3 stud speed up test setup.
Thank you for the suggestions. Yeah i having trouble connecting CD to use it as CD as shown in the picture.
It seems the pinout layout for the CD in the multisim is quiet different than the actual CD Message 2 of CD was used because it was comparable with Datashret. The wide frequency range, from 0. I am not sure how to connect each other.
Message 7 of Get Waves directly in your inbox.
【KM48C512BLJ-7 SAMSUNG】Electronic Components In Stock Suppliers in 2019【Price】【Datasheet PDF】USA
Thank you for uploading the 4538h, I have opened and compared it with the ones I found online for the CD, and they look pretty much the same, so should I assume that you have already looked at the datasheets of both components to determine if the pinout is the same?
Vibration testing and validation. Message 10 of Hope that you could find this helpful!